Please note there are many variables that can change the pressure such as tubing length, temperature, and solvent composition. Below is a guideline of the expected pressure range under normal conditions using mobile phase.
- PP-ODS for DA and 5HT - 7.5 to 10 MPa
- AC-GEL for ACh - 4.5 to 6 MPa
- CA-5ODS for NE and E - 8.5 to 11 MPa
- GU-GEL for Glutamate - 5 to 7 MPa
- SC-5ODS for Monoamines - 6 to 9 MPa
- SC-3ODS for Monoamines - 9 to 13 MPa.
- AC-GEL for Acetylcholine- 4-6 MPa
- CAX for NE, DA, 5-HT- 7 to 9 MPa
- NOPAK (ENO-20) for Nitrite and Nitrate-1.2-1.8 MPa
Please also note that these values are just for reference.
- Time to repack the precolumn.
- Time to wash/replace the separation column.
- Clogging in the tubing/connection fittings.
- Flow problem. Check the measured flow rate compared to the set value.
- Check all connecting tubing for leaks.
- Air bubbles trapped in the pump (recommend purging).