Back to website
What do you need help with?
Enter your search term here...
Submit A Ticket
Submit A Ticket
A sine wave over several minutes on the chromatogram is found during monitoring the baseline of the ECD. HTEC-500
Auto Zero: Confirming the connection between AS-700 and HTEC-500/510
Baseline wanders, drifting.
Broad peaks with a manual injection
View all (51)
Eicom Support Videos
Video Tutorial- Assembling the HTEC Pump
Video Tutorial-HTEC Piston Seal Quality Check, Replace
Video Tutorial-Disassembling HTEC Pump Head
Video Tutorial-HTEC Pump Error 3 or Error 4
Video- How to use the manual injector with the HTEC
View all (6)
A large negative peak before nitrite is interfering with the nitrite peak with biological samples, what causes this?
ENO-20 does not start. Sequencer Error
ENO-20/30 NOx Analyzer shows Error 3
ENO-20/30 Shortened Retention times for tissue or blood Samples
View all (17)
Antioxidant for catecholamines (norepinephrine, dopamine) and serotonin
What is the number on the package for RT-5S tubing used for the ERP-10 roller pump?
How to calculate dead volume of microdialysis probes
MI-AZ, MI-PEP, Injection Needle, Cannula Information
Ringer's Solution, aCSF Composition, Recipe
View all (12)
AS-700 Cooling System does not work.
Vials With Autosampler AS-700
Partial Loop vs. Microliter Pickup Injection methods
Different or large solvent front peaks with autosampler.
Glutamate/GABA OPA Derivatization Autosampler Method Description
View all (8)
How to separate a rider peak (smaller overlapping peak)
Can we move the data file folders to the desktop in Clarity?
LED Array FAQs
Voltage-Light Power Relationship for White LED
Q: What colors are available for the array?
Q: What is the output for the array?
Q: What are the dimensions of the LEDA-X array?
Q: Do you know if the light power is uniform between and within each well in the 96 well plate? Most LEDs have a cone-like spread, so I was wondering if the cone covered the full surface of cells in each well.
View all (10)
Teleopto Optogenetics FAQ
Q: Where can I find out more about how to plan my optogenetics experiment?
Q: How long will it take for me to receive my order?
Wireless Optogenetics FAQs
Q: How wide is the cone of light produced by the fiber optic implant?
Q: How close together can the two fibers of a bilateral LED implant be?
Q: This system is almost perfect for us, but would be more ideal with customizable options - can you tailor it to meet our specific needs?
Q: Which receiver should I choose, pulse type,continuous type, or OptoFlash?
Q: How long will a wireless receiver keep powering the LED before the rechargeable battery is exhausted?
View all (22)
Drinko Measurer User’s Guide & Sample Protocol
Solutions for Strex Cell Stretch Systems
Does autoclaving affect cell adhesion on Strex stretch chambers?
Can we reuse the Strex stretch chambers?
What are the materials and characteristics of the Strex stretch chambers?
Is a uniform stress load applied on every cell in the Strex stretch chamber?
What is the difference in cell response between sustained stretching and cyclic stretching?
View all (49)
Solutions for Strex Controlled Rate Freezer
What is the system capacity for cryogenic vials of the Strex Controlled Rate Freezer?
What is the Minimum Velocity of Freezing that the Strex Controlled Rate Freezer Provides?
What is the Maximum Velocity of Freezing for the Strex Controlled Rate Freezer?
What is the benefit of using a Strex Stirling Cooler system vs. a freezer that uses liquid nitrogen?
What happens if there is a power outage during use of the Strex Controlled Rate Freezer?
View all (8)
Strex User Manuals for Download
Download Strex Product Manuals
Tomy Autoclave Solutions- ES Series
No display appears on the control panel of the ES Tomy Autoclave
Steam leaks from the chamber lid of the Tomy ES autoclave
Sterilization is not performed sufficiently for the Tomy ES series autoclave
The pressure does not rise in the ES Tomy autoclave
The chamber lid cannot be opened after sterilization using the ES Tomy autoclave
Tomy Autoclave Solutions- SX Series
Tomy Autoclave SX Series Error Codes
Nothing is displayed on the Tomy SX autoclave, even with the power supply switch turned ON.
Temperature rises slowly for the Tomy SX Autoclave
Sterilization is not carried out sufficiently
Steam leaks from the chamber lid of the Tomy SX series autoclave
View all (11)
Tomy Centrifuge Solutions-MX Series
No display appears on the control panel for the Tomy MX Centrifuge
The Tomy MX Centrifuge abnormally vibrates and/or makes a loud sound.
The Tomy MX Autoclave does not start running.
The Tomy MX Autoclave does not properly cool down.
The Tomy MX Centrifuge chamber lid does not open.
View all (8)
Ask the Community
The perfect place to get insider tips from users and staff